Hold the Light Guide Column Handheld Infrared light therapy devices over the sore area at a distance of 1cm away from the skin, and keep the light in the treatment area for one minute. Do not move or wave the device across the treatment area.
If the treatment area is larger than the beam size, please separate the treatment zone into parts, and finish them one by one.
If you feel any discomfort triggered by the use of the device, increase the distance of the device from your skin or stop treatment immediately.
During the operation, aim the top at the correct position. Meanwhile, it starts counting down and stops working when counting down to “0”.
During operation, the therapy can be stopped by pressing the power button at any time.
This equipment will turn off automatically if don’t do any action within 5 minutes, turn it on by pressing any button.
After a working cycle Can press the power switch to start another working cycle. Stop operating if the equipment began to heat obviously, and do not restart until the equipment cools down. Suggest continuing the working cycle less than 5 times.
The low-power detective circuit is fixed inside the main unit, when low power is detected, the indicator light is red, please stop using it and charge.
When the battery needs to be charged, connect the plug of the adapter to the AC100V~240V power supply, then connect the output plug of the adapter to the DC 5.0V 1.0A input plug of the body. During charging, the indicator light flashes green, and when it is fully charged, the indicator light turns orange.